|Reading Time||10-15 minutes||Shelf Life||3 Years|
Individually packed test device,
|Accuracy||＞99.9%||Feature 1||High Accuracy|
|Feature 2||High Sensitivity||Feature 3||Clear Results|
|Feature 4||Easy To Read||Feature 5||Easy To Use|
|Feature 6||OEM/ODM Available||Feature 7||Quality Guarantee|
Instructions must be read entirely before taking the test. Allow the test device controls to equilibrate to room temperature for 30 minutes (20°C-30°C) prior to testing. Do not open the inner packaging until ready, it must be used in one hour if opened (Humidity: 20%~90%, Temp: 10°C-50°C) .
1. Take off the outer packing, put the cassette onto the desk with the sample window up.
2. Drop 1 drops of serum or plasma (30μl-45μl) vertically into the sample hole of cassette well, making sure that there are no air bubbles. Add about 1 drop of (35μl-50μl) sample buffer into the sample hole of cassette immediately.
3. Observe the test results immediately within 15 minutes. Don’t read result after 15 minutes. To avoid confusion, discard the test device after interpreting the result.