|Reading Time||10-15 minutes||Shelf Life||3 Years|
Individually packed test device,
|Accuracy||＞99.9%||Feature 1||High Accuracy|
|Feature 2||High Sensitivity||Feature 3||Clear Results|
|Feature 4||Easy To Read||Feature 5||Easy To Use|
|Feature 6||OEM/ODM Available||Feature 7||Quality Guarantee|
Allow the test device and specimens to equilibrate to temperature (15-30°C or 59-86°F) prior to testing.
1. Bring the pouch to room temperature before opening it. Remove the test cassette from the sealed pouch and use it as soon as possible.
2. Place the test device on a clean and level surface.
3. Hold the dropper vertically and transfer 1 drop of specimen (approximately 35μl) to the specimen well(S) of the test device, then add 2 drops of buffer (approximately 70μl) and start the timer. See illustration below.
4. Wait for the colored line(s) to appear. Read results at 15 minutes. Do not interpret the result after 20 minutes.