|Reading Time||10-15 minutes||Shelf Life||3 Years|
Individually packed test device,
|Accuracy||＞99.9%||Feature 1||High Accuracy|
|Feature 2||High Sensitivity||Feature 3||Clear Results|
|Feature 4||Easy To Read||Feature 5||Easy To Use|
|Feature 6||OEM/ODM Available||Feature 7||Quality Guarantee|
Allow the test device and specimens to equilibrate to temperature (15-30°C or 59-86°F) prior to testing.
1.Remove the test cassette/strip from the sealed pouch and use it as soon as possible.
2.Place the test cassette/strip on a clean and level surface.
3.For serum or plasma specimen: Hold the dropper vertically and transfer 3 drops of serum or plasma (approximately 100 μL) to the specimen well (S) or specimen pad of the test cassette/strip, then start the timer. See illustration below.
4.For whole blood specimens: Hold the dropper vertically and transfer 1
drop of whole blood (approximately 35 μL) to the specimen well (S) or specimen pad of the test cassette/strip, then add 2 drops of buffer (approximately 70 μL) and start the timer. See illustration below.
5.Wait for the colored line(s) to appear. Read results at 15 minutes. Do not interpret the result after 20 minutes.