A. Samples processing and control:
① Place the sample processing tubes on the worktable, and add 6 drops of extraction solution A(approximately 390μl).
② Put the sampling swab into the sample processing tube, which contains extraction solution A, and store at room temperature, and continuously rotate and squeeze the swab in the tube wall in the process, so that the liquid is constantly being squeezed out, and repeat it for several times and deal with it for 2 minutes.
③ Then add 6 drops of extraction solution B (approximately 390μl), and rotate and squeeze swab, as far as possible make the liquid flow out, and then discard the swab in line with the handling method of infected goods.
④ Fit the dropper tip on top of the sample processing tube.
⑤ If the samples shall be used, within 60 minutes after treated, this will not affect detection results of the kit.
Notes: The dropping amount of solution A and solution B should be equal, when the swabs is processed.
B. Detection steps:
① Please read the instructions for the test kit carefully before operation.
② Take out the test kit from the sealed pouch, and place it in a clean, dry and level worktable. If the test kit is stored at the place, which temperature is lower than room temperature, and the test kits and reagents required to be taken out in advance, and it should revert to room temperature before it is being used.
③ 3 drops of processed samples in the sample processing tube should be dripped into the sampling wells of the test kit.
④ Wait for the appearance of the results. 10 minutes after the sample is dripped, the results can be interpreted. The time of the red line varies in line with the contents of Gonorrhea, which is collected with swabs, and some positive samples may show the result after 60 seconds. In order to ensure the negative results, please do not judge the result after 15 minutes.